INTRODUCTION SCION 456-GC Configuration
Low level detection of permanent gases and carbon dioxide is essential for Ultra High Purity helium since traces of these components affect the quality and thus the value of the helium gas. The quantification of trace gases is highly applicable using Gas Chromatography (GC) due to the complete separation of the analytes prior to detection, and in addition, the non-sensitivity of the Pulsed Discharge Helium Ionization Detector (PDHID) to helium.
Performing a complete separation of permanent gases and carbon dioxide by GC is not possible using a single column. The common solution is a configuration of two identical analytical channels with the two columns having a different type of stationary phase. By optimisation of the column dimensions, it is possible to combine the two columns into one channel. This application note describes the trace analysis of permanent gases and carbon dioxide in helium using a single channel configuration with two columns connected in parallel.
EXPERIMENTAL SCION 456-GC Configuration
The SCION 456-GC configuration consisted of two parallel columns for the optimisation of a fast separation of permanent gases. The sample is injected by a gas sampling valve via a mini-gas splitter to two columns, as illustrated in Figure 1.
A SCION-Molsieve 5A column is used to separate hydrogen, oxygen, nitrogen, methane and carbon monoxide before the composite air peak elutes from the SCION PLOT-Q column. After this composite peak, the methane and carbon dioxide elute from the SCION PLOT-Q column. Column choice was optimised to ensure that all permanent gases elute before the t0 time of column 1, preventing any mixed elution from both columns.
Table 1 shows the analytical conditions of the SCION GC.
When using PDHID as a detection technique with the parallel column setup, the permanent gases and carbon dioxide can be analysed down to a lowest detection limit (LDL) of approximately 50ppb within 120 seconds. With the high inertness of the SCION molsieve column, all components including carbon monoxide elute with a perfect peak shape allowing low detection levels, as shown in Figure 2.
As methane elutes from both columns it was used as a component to validate the analytical performance, between the two columns. The repeatability results are all less than 1.0% RSD, over 20 consecutive GC runs. To validate the concept of parallel column analysis, the repeatability of methane from both columns is used. The repeatability of methane from the individual columns (Molsieve RSD 0.65%, PLOT RSD 0.72%) is very similar to the repeatability of the summed methane (RSD 0.66%). Figure 3 shows the percentage of methane injected on the two columns. The amount of methane entering the Molsieve column is 67% with 33% entering the PLOT column.
Table 2 shows the mean, standard deviation and relative standard deviation (RSD%) of low level trace gases in the helium, at the 2ppm level with 20 injections. The excellent repeatability proves the working concept of parallel columns where the sample is split between two columns.
The SCION 456 GC configured with parallel column setup with the use of the PDHID detector ensures trace levels of permanent gases and carbon dioxide are analysed within 120 seconds, down to an LDL level of 50ppb. The validity of the instrumentation was shown through exceptional repeatability when parallel columns with a single injector and detector system is used for high purity helium analysis.
Sriracha mi artisan farm-to-table bushwick health goth, fixie palo santo adaptogen. Affogato kale chips butcher shoreditch gastropub palo santo. Sartorial selvage sriracha polaroid forage umami. Sriracha locavore poke occupy, post-ironic quinoa live-edge chartreuse shabby chic ugh williamsburg cray everyday carry franzen.
Charcoal bottle listicle cred portland lomo, bicycle rights unicorn swag truffaut cold-pressed squid cloud bread quinoa. Activated charcoal keffiyeh sriracha lumberseual, poke meggings hashtag synth direct trade. Single-origin coffee skateboard literally occupy sema gochujang swag authentic bit.